I am having issues getting the correct processed Accel. data from the UM7 as was shown in the videos found on the product page.
I followed most of the steps shown in the videos found on the product webpage however when I go to graph the Processed Accel data using the CHR Serial Interface instead of the getting x=0, y=0 and z = -1, I have y= 0, z=-1 and x = -2.
Is there some additional calibration or other procedure, that I am supposed to be doing or is there something wrong with the IMU chip itself?
That’s definitely odd. I haven’t observed that behavior before. Can you open the data plot, make sure it is the active window, and take a screenshot? Use Alt-PrintScreen to just take a picture of the data graph. You can then paste it into an email and send it to firstname.lastname@example.org, or upload it to this forum.
Viewing 2 posts - 1 through 2 (of 2 total)
You must be logged in to reply to this topic.
Thank you very much for your patronage. We are experiencing production and shipment delays due to the coronavirus infection that is currently occurring. We apologize for any inconvenience this may cause. We will work hard to deliver the product as soon as possible. Dismiss